Release Notes – guavaSoft™ Software version 2.7 Introduction This document contains information about release 2.7 of EMD Millipore Corporation guavaSoft Software for your EMD Millipore Corporation easyCyte™ or easyCyte HT System. The contents of this document are:
Summary of New Features, Changes and Enhancements since guavaSoft Software v2.6
Installing guavaSoft Software v2.7
Using guavaSoft Software v2.7
Customer Support Contact Information
Appendix A. Analyzing EMD Millipore Corporation easyCyte HT System Data Using FCS 2.0 File Export
Important Notices guavaSoft Software v2.7 is intended for use only on EMD Millipore Corporation easyCyte or easyCyte HT Systems. Do not install guavaSoft Software v2.7 on a Guava® PCA, Guava PCA-96, Guava PCA-96 AFP, Guava EasyCyte Mini, Guava EasyCyte, Guava EasyCyte Plus system computer for acquisition purposes.
To run guavaSoft Software v2.7 for acquisition purposes, you will need an EMD Millipore Corporation-configured Dell™ laptop running 32-bit Windows XP Professional or 32-bit Windows 7 Ultimate, and including Microsoft Excel®. (Minimum specifications: Intel Core Duo® processor T2400 (1.86 GHz), 4 GB RAM, 60 GB hard drive, 2 USB ports, CD or DVD drive, full screen display resolution of 1024 x 768 or wide screen display resolution 1280 x 800).
Summary of New Features, Changes and Enhancements since guavaSoft Software v2.6 Guava Incyte™SoftwareModule - New Features
Support was added for Access Rights management (similar to that in older Guava assays), to aid compliance with 21 CFR 11 regulations.
A new interactive batch printing feature allows easy selection of plots and samples per page, with preview and the ability to save to a PDF file.
Many improvements to table printouts: better layout, number alignment, and other features to save paper.
Heatmap Stats: You can now drag a stat from the Current Run Stats table to one of the sectors in the Pie Legend for the heatmap.
Zoom and Pan for plots.
Contour plots, with adjustable levels and smoothing.
Histogram smoothing, with adjustable smoothing.
New free-form polygon region (in addition to the existing octagonal polygon region).
EC-50/IC-50 plots: you can now copy and paste concentration values back and forth between the EC-50/IC-50 Setup and an Excel spreadsheet.
Maximum number of plot was increased to 24 (added 4x4, 4x5, 5x4 and 5x5 layouts).
Maximum number of regions was increased to 24 (from 16 previously).
When you delete an Analysed Group, the data group and Analysis Method it uses are also deleted automatically if no other group is using them.
Fixed a crash when trying to re-import an FCS 2.0 file which had been previousy exported.
Guava Incyte™Software Module - Issues Addressed
Addressed item: A larger number of events may sometimes be indicated for some channels during acquisition of easyCheck replicates.
Installing guavaSoft Software v2.7 To install guavaSoft Software v2.7, perform the following steps.
Log into the computer with Administrative rights. This is the default at installation; consult your local system administrator if necessary.
If you have been using a previous version (guavaSoft Software v1.0, guavaSoft Software v1.1, guavaSoft Software v2.0, guavaSoft Software v2.1, guavaSoft Software v2.2.3 or guavaSoft Software v2.6), the previous version will be automatically uninstalled (the easyCheck log and event log files will be saved and copied to the new installation). In case Installation issues are encountered, please use the Windows Control Panel to uninstall the older guavaSoft program
Insert the CD-ROM into the computer and open its contents, where you will see a folder labeled "guavaSoft 2.7". You can use the contents of this folder to install the software on the laptop that is connected to the instrument. You may also install the software for off-line data analysis on any computer running, at a minimum, Windows XP; however, note the disclaimer at the end of this document.
Open the folder and double-click on the Setup.exe file to start up the installer.
You should then see a window titled "Welcome to the InstallShield Wizard" (in some cases, this window may be hidden behind other windows, so you may need to use the Windows task bar to bring it to the front). The InstallShield® Wizard will lead you through a series of screens for the various installation steps. Simply follow the instructions and accept the default settings by clicking the Next button. Note the following, however:
When the "License Agreement" window appears, EMD Millipore Corporation recommends that you read it carefully in case it has changed since the previous release. You must select the radio button that indicates your acceptance of the license terms before you can click on the Next button to continue.
If the "Customer Information" window appears, be sure to select the option to "Install this application for anyone who uses this computer (all users)," then click Next. Do not select the option to "Install the software Only for me" as the software may not install correctly.
When installing the software for off-line data analysis on a computer not attached to an instrument, select “Skip” on the “Instrument detection” window in order to access the “Setup Type” window. Select the instrument type by clicking on the appropriate radio button.
After the software has been installed and the last screen appears, click on the "Finish" button to complete the process. You are now ready to use the new software.
Using guavaSoft Software v2.7 Please refer also to the User's Guide appropriate for your system which can be found on the software distribution CD-ROM. The following notes apply to all users of guavaSoft Software:
Running Excel, Internet Explorer, or any other program while using guavaSoft Software to acquire data from your system is not recommended. guavaSoft Software requires the full resources of your laptop during data acquisition; running other programs at the same time (even if you are not actively using them) may interfere with its ability to acquire data reliably.
When acquiring data with guavaSoft Software, always store the data onto the laptop's local hard drive. Do not use a remote file server on a network, as network connections are often intermittent or busy, and this may interfere with the software's ability to acquire data reliably. If you want to archive data to a remote file server, wait until the session is finished. Then exit the assay (i.e., return to the guavaSoft Software main menu), and copy or move the data to a remote file server.
For best performance, EMD Millipore Corporation recommends that you use the Medium flow rate, except for the Guava Cell Cycle Software Module, which uses Low or Very Low flow rates only. The Guava ViaCount, Guava Check, and guava easyCheck Software Modules allow samples to be run only at the Medium flow rate.
Always make sure the power is off for both the instrument and the laptop before inserting or removing the USB cable, otherwise damage or erratic operation may result.
EMD Millipore Corporation recommends that you re-boot the laptop at least once each day, e.g., in the morning before beginning to use your system.
In order to view the Access Control screen, you must be a Guava Administrator. In order to change the settings, however, you must also be logged in a Windows administrator.
Setting the threshold too close to a population may cause some events close to the threshold to be rejected as noise.
In Analysis Mode, if you enter new values into the Dilution Factor or Total Volume fields in the ViaCount, ExpressPlus and ExpressPro Software Modules, the changes will not register until you click on a different item.
In Analysis mode, the ExpressPlus Software Module v3.6 (which is included in this installation) can convert files acquired in Express Plus Software Module v3.4 and earlier versions. In addition, files generated with guavaSoft Software v2.1 are compatible with earlier versions of software for the EasyCyte System (versions 3.0 and later), Guava PCA software (versions 2.1 and later), and Guava PCA-96 software (versions 2.5 and later) for ViaCount, Nexin, MultiCaspase, TUNEL and Cell Cycle. guavaSoft Software v2.2will also correctly display and analyze data acquired on the PCA-96 or PCA even for the CellPaint, and CellToxicity Software Modules in which the fluorescence parameters used are different between CytoSoft Software v5.3 and v2.5.7 or v6.0.2
On rare occasions, the software and the instrument may pause to resynchronize with each other. If this occurs, the system will pause for 20-30 seconds and a message will appear in the status bar at the bottom of the screen indicating a “Tray Hold-Off.” Although this is not serious and the system should resume again after resynchronizing, be aware that after the tray hold-off occurs, the tray may move without warning.
Unplugging a network cable when a network printer is installed and selected as the default printer will result in CytoSoft and guavaSoft Software Modules being unable to launch. This problem only occurs on laptops. An error message will appear stating that the application has generated errors and will be closed by windows. A workaround is to select "Fax" as the default printer.
All Errors and Warnings are noted in the Event Log. All Errors and Warnings that occur while a sample is being acquired will also be posted to the Errors and Warnings counter located at the bottom right of the acquisition screen and added to the appropriate column in the CSV file (for ExpressPlus and ExpressPro only). Errors and Warnings that occur after a sample acquisition has been closed will not be posted to the CSV file and may or may not be added to the Errors and Warnings counter located at the bottom right of the acquisition screen, depending on the type of error or warning.
The position of the histogram markers should be used to define populations and not the color of the events included in the marker. The color indicating events included in a particular marker can be offset by 1 or 2 pixels to the left of the marker ends and hence does not quite correctly indicate which events are included in the marker.
When in batch printing mode, even if double sided printing is selected through the “properties” button on the print dialog window, the printout will be single-sided only.
Disclaimers Due to the wide variety of PC configurations, EMD Millipore Corporation does not warrant the use of guavaSoft Software to obtain results when used on platforms other than the supported laptop.
Export to the FCS 2.0 data format is provided for the convenience of our customers (see Appendix A). However, EMD Millipore Corporation has not validated the analysis of data from the system using third-party programs and cannot warrant that the results will be correct, nor does EMD Millipore Corporation provide any support for such analysis.
Customer Support Contact Information For ordering information or technical support:
http://www.millipore.com/reply/form/techservice US and Canada
Malaysia: (603)79571322; firstname.lastname@example.org For additional contact information, visit www.millipore.com For ordering information or technical service for customers in academic or government settings or research institutions, contact your nearest Millipore office. To find the office nearest you, go to www.millipore.com/offices . For technical service, go to www.millipore.com/techservice.
In addition, Millipore and its distribution network will provide Guava® Products to all sectors of life science research in certain countries outside of North America and Europe. To find the office nearest you, go to www.millipore.com/offices
Field service for all EMD Millipore Corporation’ instrumentation for all the geographies above will be carried out by EMD Millipore Corporation (www.millipore.com/techservice).